We have 6 Fresenius Medical Care K manuals available for free PDF download: Troubleshooting Manual, Operator’s Manual, Technical Manual. View and Download Fresenius Medical Care K operator’s manual online. K Hemodialysis Machine Operator’s Manual. K pdf manual download. FRESENIUS K OPERATORS MANUAL KHOXJMBJNW | PDF | 45 Pages | KB | 06 Apr FRESENIUS K OPERATORS MANUAL.
|Published (Last):||16 May 2014|
|PDF File Size:||5.10 Mb|
|ePub File Size:||2.67 Mb|
|Price:||Free* [*Free Regsitration Required]|
If unable to reset the alarm, call your local qualified technician. Moving The Manial The K can upload treatment information to a network database for review by clinical staff using a personal computer.
With some RO systems, the water inlet line may be disinfected along with the RO and distribution piping by leaving the dialysis machine in Rinse manuwl during RO disinfection. Dialysate should already be verified for absence of disinfectant, verification of prescription, conductivity, and pH should also be confirmed.
Page 84 The drip chamber and transducer are connected via frewenius pressure line that runs from the chamber to venous pressure port P located on the front of the module. The end time is represented in the profile graph by a tick mark on the horizontal axis.
The yellow dotted line shows the alert level.
New Fresenius k semi & annual pm manual – Topic
Function Of The k If alarm is not cleared, take the machine out of service. Page If the message is not Message cleared, alert a qualified technician.
Manually Operating the Blood Pump In the event of a power failure during treatment, the K blood pump can be manually operated to return the blood to the patient or mnual keep the blood in recirculation if a quick resumption of power is anticipated. Page This option defines whether or not a Diasafe Plus filter is present in the machine. Enter text from picture: Pre-uf Pump Filter Measuring Fluid Volumes Fresenijs the machine uses a closed, volumetric ultrafiltration system, the TMP is monitored primarily for detecting large shifts in pressure.
Page 11 These instructions are for the Fresenius K 2008, machine. These screens are the: Rebuilding The Ultrafiltration Pump Other companies seem to be able to put out new equipment in the USA.
Machine alarms may not occur in every blood loss situation. Concentrate And Bicarb Pumps When prompted, connect red connector to chemical disinfectant.
Page Machine alarms may not occur in every blood loss situation 5. Annual Hour Preventive Maintenance Page Company neither assumes, nor authorizes the assumption of, any obligation or liability in connection with such equipment.
Fresenius Medical Care 2008K Operator’s Manual
The battery cartridge will pop forward. Test Check UF pump. Chapter 1—Overview Modules The modules accompanying the K are located just below the control panel. Check all bloodlines and dialysate lines for leaks. Preparing The Dialysis Delivery System Page TP that is to be replaced with each treatment. Page Wait until blood pressure module completes resetting and retry blood pressure reading.
Daily Preparation For Treatment Locate the frwsenius on the back of the machine and push the black battery loading cartridge in and to the left.